New year's holiday notice in Mikrouna
Winter vacation is coming soon, it is very important to do the maintenance of the glove box in advance!
The students are about to have winter vacation. In order to make the glove box in our laboratory work better during the winter vacation, the glove box will be in a more stable state next year. You need to do the following!
01. Replace a full bottle of working gas to prevent the working gas from running out during the holiday and not being replaced in time;
02. Exhaust negative pressure for large and small warehouses to prevent the hatch cover from not being tightly closed, causing the cabinet to leak;
03. The cycle is normally open during the holidays. The laboratory temperature is higher than 15 degrees Celsius. It is recommended that the cabinet pressure be set to upper pressure 5 and lower pressure 1. When the laboratory temperature is lower than 15 degrees Celsius, it is recommended that the chamber pressure be set to upper pressure 4 and lower pressure. 0. This setting is to prevent thermal expansion and contraction. If the laboratory temperature is lower and the pressure setting is higher, it will consume more working gas;
04. If there is an open solvent in the glove box, be sure to seal the solvent before leaving on holiday to prevent the solvent from volatilizing and affecting the performance of the glove box;
05. The more precious and active metals are recommended to be sealed and preserved before holidays to prevent the circulation of working gas from being insufficient, which will cause the water and oxygen in the box to rise and damage the samples;
06. Check whether the vacuum pump oil is clean, if it is not clean, replace the pump oil in advance;
07. It is recommended to turn off the analyzer when no one uses it during the holiday, and then turn it on after the holiday ends;
08. Do not cut off the power to the main switch of the glove box. If there are electrical equipment in the box, the power should be cut off before leaving for holiday;
The above suggestions have played a good protective effect on the maintenance of the glove box. Maybe you are still a little worried, because you can’t see the real-time running status of the equipment like every day. In order to solve your worries, through our Internet of Things System, you can achieve the following functions:
1. Real-time data query, for example, you can view the real-time data of your glove box water and oxygen value, box pressure real-time value, upper and lower pressure setting value and other real-time data through your mobile phone at home;
2. Real-time control function, such as you forget to turn off the lighting when you leave the laboratory, you can remotely turn off the lighting through your mobile phone;
3. Setting function, you can modify the upper and lower pressure setting value of the glove box according to the change of weather;
4. Cleaning function. For example, the water and oxygen level in the glove box has increased due to the tightness of the items. You can turn on the cleaning function of the glove box through your mobile phone at any time to reduce the water and oxygen level to the normal range in time;
5. Historical data chart display function, you can view the change curve of historical data at any time to understand the change trend of equipment data;
6. Historical data report function, you can download the core data report to view all data of a certain experiment or a certain period of time;
7. Alarm information push function, when the glove box is powered off or caused by other reasons, the alarm information will be sent to you in time by SMS and WeChat;
8. Device centralized management function, if you have multiple devices, you can manage each device separately through the system.
The entrance to the MI Cairo IoT system experience
(You can click the demo account in the lower left corner of the login page to enter the system)